Date: Sat, 04 May 2002 12:40:54 -0400 (EDT)
Author: Steve Wonnell
Subject: Re: Yet Another 'Scope Question
I don't have the scope in front of me, but if you are triggering the scope
(single shot or normal modes), you might try pressing the run/stop button
once or twice (i.e. to trigger it manually). Another possibility, once
you have the scope set up the way you want it, is to save the settings
(not the waveform) into memory. After runing the magnet through the coil,
you can recall the setup to restore the original settings, which should
put you back where you started.
On Sat, 4 May 2002, George M. Caplan wrote:
> Subject: Yet Another 'Scope Question
> I am (slowly) learning how to use the Tek TDS-210 'scope.
> How do I blank the display w/o powering down?
> If I have a waveform displayed, it will become dimmer if I adjust
> the trigger level or make any other change that would affect
> its interpretation. But how do I actually get rid of the display.
> I am using the 'scope to display the waveform in a magnetic induction lab.
> I run a magnet through a coil & show the students the resulting
> waveform. After the first time, the screen is not blank when I
> start. This is confusing to the students (& to me, too).